Different lead locations guided by fluoroscopy and ECG parameters and their effect on patients functional status
The Egyptian Heart Journal volume 71, Article number: 29 (2019)
Right ventricular apical pacing with the resultant left ventricular dyssynchrony often leads to depressed systolic function and heart failure. This study aimed at investigating the relation between various septal locations guided by ECG and fluoroscopy and the intermediate term functional capacity of the patients.
Fifty patients who received a single lead pacemaker with assumed > 90% pacemaker dependency. Patients were randomized according to RV pacing site RV into group 1 “high septum” (n = 15), group 2 “mid septum” (n = 25), and group 3 “low septum” (n = 10) using QRS vector and duration as well as fluoroscopic parameters. Their clinical status was assessed 6 months after device implementation using 6-min walk test (6MWT).
The study showed that paced QRS complex duration itself has no significant difference between the different septal pacing locations (P-value 0.675), although its combination with the paced QRS complex vector can signify the optimal pacing site and 6MWT showed a significant difference among the groups in favor of group 1; group 1 (413.3 ± 148.5), group 2 (359.8 ± 124.6), and group 3 (276.0 ± 98.5) P value 0.04.
There was a significant difference found between the three septal pacing sites concerning the patient functional capacity with superiority of high septal location. By contrast, different septal sites showed no significant difference of the paced QRS complex duration. To optimize the pacing site in the septum, assessment of the paced QRS vector in leads I and III is of a great benefit especially when combined with paced QRS complex duration assessment.
Right ventricular (RV) apical pacing allows for safe and stable long-term pacing. However, it induces non-physiologic left ventricular (LV) activation with alteration of the intraventricular contraction sequence, which delays LV activation . This delay is accompanied with LV dyssynchrony, and the development of LV dyssynchrony was reported to be associated with deterioration of heart failure symptoms and the systolic LV dysfunction .
To overcome this potential limitation, other pacing sites throughout the whole septal segments have been tried . The septal areas, particularly the mid-right ventricular (RV) septum and the RV outflow tract (RVOT), have been proposed as alternative pacing sites to RV apical pacing leading to a more physiologic electrical conduction to the left ventricle (LV) and therefore to a more physiologic contraction .
The degree of electrical dyssynchrony induced by RV pacing can be estimated by the duration of the paced QRS. Moreover, some studies have demonstrated that patients with a long paced QRS may be at high risk of adverse cardiac events including heart failure. A long paced QRS can result despite the non-apical placement because the lead might pace a non-favorable septal segment .
The aim of our study is to investigate the relation between various septal locations guided by ECG and fluoroscopy and the intermediate term functional capacity of the patients through illustrating the relationship between different septal positions and paced QRS complex duration and the relationship between the paced QRS vector and duration of the paced QRS complex.
This cross-sectional study conducted during the period from October 2017 to July 2019 at our cardiology department on 50 patients having a primary implant single lead VVI pacemaker following the recent guidelines for cardiac pacing with high probability of device dependency (more than 90%) excluding those with significant valve disease (i.e., mitral insufficiency 75% and worse, moderate or severe aortic stenosis), recent (within 3 months) acute coronary syndrome, planned cardiac surgery (coronary artery bypass grafting, valve surgery), and ejection fraction of left ventricle less than 50%.
A detailed informed verbal consent has been taken from all participants as it was just an observational study and this consent was approved by our ethics committee.
Fifty patients were divided into three groups depending on the septal location with fluoroscopy guidance: group 1 high septum, group 2 mid septum, and group 3 low septum.
All participants were subjected to a follow up visits for 6 months after implantation of the device. A detailed history taking, physical examination, ECG data (including assessment of paced QRS complex duration and vector configuration), and assessment of their functional status using 6MWT were done in our cardiology department.
Fluoroscopic definition of septal pacing sites
To facilitate lead placement in the RV septum, a 3 × 3 square grid is created for each patient using a fluoroscopic image in the right anterior oblique (RAO) 10° view using landmarks to demarcate the borders of the grid as shown in (Fig. 1) that facilitate to clarify the high, mid, and low septal locations. After fixation of the septal lead using the RAO view, the left anterior oblique (LAO) view (40–45°) is used to confirm that the lead is on the septum and not the free wall (Fig. 2) 
Determining the paced QRS complex duration
All ECG records are made using an identical ECG device (Nihon Kohden Cardiofax C ECG-1150. Japan) of limb leads I, II, and III at speed 50 mm/s at a ventricular rate less than 80 bpm. The paced QRS complex duration is calculated from the end of the pacing spike to the latest deflection of the QRS complex in any limb lead. The non-paced QRS complex duration is measured from the earliest deflection to the latest deflection of the QRS complex in any limb leads. Optimization of the pacing site of the RV septum occurred by the maximal shortening of the QRS complex duration.
Defining the paced QRS complex vector
Assessment of the paced QRS complex vector is simplified to determine the R/S voltage ratio in leads I and III. Three categories are defined positive, isoelectric, and negative. A positive vector is determined as R-S ratio > 1/2 R. An isoelectric vector is determined as R-S ratio < 1/2 R or S-R ratio < 1/2 S. A negative vector is defined as S-R ratio > 1/2 S. Then, they are divided into three subgroups: (1) negative or isoelectric in lead I plus positive in lead III. (2) Isoelectric or negative in lead III. (3) Positive in lead I plus positive in lead III .
Standard descriptive statistics were used for the analysis; continuous parameters of QRS duration were described by means and standard deviation, while occurrences of categorical parameters (gender, pacing site) were described by count and percentages.
Differences between parameters of QRS duration and vector according to several factors and their combination were assessed by ANOVA and also by chi-square tests. For the studied groups, statistically significant differences at the level of significance of 0.05 are stated.
Post hoc tests analysis was done over the paced QRS complex vector result and also the 6MWT result to compare each group with the other group specifically.
For data analysis, the SPSS Statistics 17.0 for Windows (SPSS Inc., Chicago, IL, USA) was used and α = 0.05 was considered as the level of statistical significance in all analyses.
Fifty patients were divided according to the RV lead pacing site into group 1 “high septum” (n = 15), group 2 “mid septum” (n = 25), and group 3 “low septum” (n = 10). There are three patients who were excluded through the follow-up period and the reasons were death, ischemic cerebral stroke, and non-compliance. No other complications have been reported.
The demographic characteristics of the 50 patients are shown in (Table 1). Risk factors were adjusted in the three groups to eliminate interference of risk factors on the results and revealed no significant difference between the groups.
All patients were subjected to vector configuration using leads I and III and classified into to three configurations accordingly: A “negative or isoelectric in lead I and positive in lead III” (n = 19), B “isoelectric or negative in lead III” (n = 15), and C “positive in lead I and positive in lead III” (n = 16).
Patients’ clinical status was assessed after VVI pacemaker implementation within the follow-up period using 6-min walk test (6MWT) counting the meters the patient can walk freely through the examination time and taking its all precautions in consideration.
The septal position of the pacing lead in the right ventricle is a preferred alternative pacing site given the potential contribution of limiting LV dyssynchrony compared to RV apex stimulation. Historically, the decision of where in the septum the lead should be placed is individual, with the selection being corrected according to the paced QRS complex duration .
In our study, 50 patients were divided according to the RV lead pacing site by new documented fluoroscopic method into group 1 “high septum” (n = 15), group 2 “mid septum” (n = 25), and group 3 “low septum” (n = 10), all of them were subjected to follow-up for 6 months after VVI pacemaker implantation.
Risk factors were adjusted in the three groups to eliminate interference of risk factors on the results (Age, gender, smoking, HTN, DM, and dyslipidemia) with no significant difference between the groups.
According to the functional capacity assessment, there is significant difference that was found between the three septal pacing site (high, mid, and low) with significant superiority of high septal location.
By contrast, different septal sites defined by the mentioned fluoroscopic method in the RAO 10° view showed no significant differences in paced QRS complex duration. But, assessment of the paced QRS vector in leads I and III is of a great benefit especially when combined with paced QRS complex duration assessment as group C of the QRS vector configuration showed significant longest QRS duration.
According to the paced QRS complex duration, there is no significant difference among the groups concerning the paced QRS duration, which was in agreement with  whose RCT evaluated paced QRS duration in various septal positions in a total of 609 patients where the RV lead was allocated using fluoroscopic guidance and stressed on the superiority of QRS vector configuration above the QRS duration in septal lead insertion.
And we disagree with another randomized controlled trial by  showed that RVOT septal pacing is associated with shorter QRS duration than elsewhere in the RV outside the His bundle. This suggests that pacing from the septal RVOT, although not as good as intrinsic conduction, may be the most desirable site for chronic RV pacing as a narrow QRS duration is associated with improved LV dynamics.
Another multi-center prospective study comparing the ventricular dyssynchrony according to the position of right ventricular pacing electrode  where pacemaker leads were inserted through the subclavian vein using standard implantation techniques. The RV leads were positioned in the RV apex (n = 45) or interventricular septum (n = 34) under fluoroscopic guidance. The results showed that the QRS duration was significantly increased in both groups after pacing, but the difference between the pre- and post-pacing QRS duration was significantly higher in apical pacing group (57.1 ± 28.3 versus 32.8 ± 40.5 ms). This study concluded that the apical pacing has a higher probability of more LV dyssynchrony after PM implantation.
Regarding the QRS vector configuration our study show high statistical significant difference among the groups in favor of group I high septum which goes with the result of a meta-analysis by ([8, 9] which both concluded that high septum and RVOT have been suggested as more physiological sites for cardiac Pacing and the ECG confirmation of pacing in the right ventricular septum is manifested by a negative QRS morphology in lead I, whereas pacing in the right ventricular free wall manifests as a positive QRS morphology in lead I. Also, ventricular pacing in a high position will result in an upright QRS in aVF, whereas a lower position will have a less positive QRS deflection in aVF.
A prospective observational study studied the optimal pacing site in the right ventricular septum . Overall, 304 measurements of paced QRS complex characteristics in different RVS sites were performed in (100/102 patients). This study concluded that leads I and II together with fluoroscopy view can provide useful information for selecting the optimal pacing site corresponding to the QRS duration shorter than 160 ms as a cut-off value for HF prediction (Tables 2, 3, 4, and 5).
In a total of 50 patients, there is significant difference was found between the three septal pacing site (high, mid, and low) concerning the patient functional capacity with superiority of high septal location.
By contrast, different septal sites defined by the mentioned fluoroscopic criteria in the RAO 10° view showed no significant difference of paced QRS complex duration.
To optimize the pacing site in the septum, assessment of the paced QRS vector in leads I and III is of a great benefit especially when combined with paced QRS complex duration as when the vector configuration was “positive in lead I and positive in lead III,” it showed significant longest QRS duration.
Availability of data and materials
Available in my institute with a case rate of 3–4 cases per month. Available in Cardiology Department, Faculty of Medicine, Zagazig University, Zagazig, Egypt, with a case rate of 3–4 cases per month.
Six-minute walk test
Randomized control trial
Right ventricular outflow tract
Daggett WM, Bianco JA, Powell WJ Jr, Austen WG (1970) Relative contributions of the atrial systoleventricular systole interval and of patterns of ventricular activation to ventricular function during electrical pacing of the dog heart. Circ Res 27:69–79
Tops LF, Schalij MJ, Holman ER, van Erven L, van der Wall EE, Bax JJ (2006) Right ventricular pacing can induce ventricular dyssynchrony in patients with atrial fibrillation after atrioventricular node ablation. J Am CollCardiol 48:1642–1648
Manolis AS (2006) The deleterious consequences of right ventricular apical pacing: time to seek alternate site pacing. Pacing Clin Electrophysiol 29:298–315
Mond HG, Hillock RJ, Stevenson IH, McGavigan AD (2007) The right ventricular outflow tract: the road to septalpacing. Pacing Clin Electrophysiol 30:482–491
Victor F, Leclercq C, Mabo P, Pavin D, Deviller A, de Place C, Pezard P, Victor J, Daubert C (1999) Optimal right ventricular pacing site in chronically implanted patients: a prospective randomized crossover comparison of apical and outflow tract pacing. J Am Coll Cardiol 33:311–316
Gerry K, Bruce S, Stambler BS, Raymond Y (2009) Search for the Optimal Right Ventricular Pacing Site. Pacing Clin Electrophysiol 32:426–433
Libor K, Pavel S, Tomas H, Eva K, Jiri J, Simona L, Milos T (2013) Optimization of the pacing site in ventricular septum by fluoroscopy and morphology of the paced QRS complex duration. Biomed Pap Med Fac Univ Palacky Olomouc Czech Repub 157(1):56–63
Hillock RJ, Mond HG (2011) Pacing the right ventricular outflow tract septum: time to embrace the future. Europace 14(1):28–35
Lieberman R, Grenz D, Harry G, Micheal D (2004) Selective Site Pacing: Defining and Reaching the Selected Site. Pacing Clin Electrophysiol 27:883–886
Gianni P, Francesco Z, Enrico B, Gianluca R, Giorgio C, Alberto M, Franco N, Loris R (2013) How can we identify the optimal pacing site in the right ventricular septum? A simplified method applicable during the standard implanting procedure. Am J Cardiovasc Dis 3(4):264–272
Cho G-Y, Kim M-J, Park J-H, Kim H-S, Youn HJ, Kim K-H, Song J-K (2011). Comparison of Ventricular Dyssynchrony According to the Position of Right Ventricular Pacing Electrode: A Multi-Center Prospective Echocardiographic Study. J Cardiovasc Ultrasound 19(1):15
This work was supported by Cardiology Department in Zagazig University Hospitals and by my lovely family.
Ethics committee’s reference number
No funding resources.
Ethics approval and consent to participate
This study was approved by the ethics committee of institutional review board (IRB) for medical research ethics. Faculty of Medicine, Zagazig University. A detailed informed verbal consent has been taken from all participants as it was just observational study with no intervention, and they have to implant the pacemaker by any way with our follow-up that was of a great value for them, and this consent was approved by the ethics committee.
Consent for publication
Available in the same written consent
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
MesbahTahaHassanin, M., ShafieAmmar, A., Abdullah, R.M. et al. Different lead locations guided by fluoroscopy and ECG parameters and their effect on patients functional status. Egypt Heart J 71, 29 (2019). https://doi.org/10.1186/s43044-019-0023-1